European Commission, Joint Research Centre (JRC), Institute for Environment and Sustainability, Via E. Fermi, I-21027 Ispra, Italy
E-mail: email@example.com; Fax: +39 0332 789931
; Tel: +39 0332 785652
J. Environ. Monit., 2012,14, 2145-2153
03 Feb 2012,
12 Jun 2012
First published online
13 Jun 2012
Monitoring personal exposure to particle matter (PM2.5) in ambient air requires performing measurements using portable monitors. In this work, the portable nephelometer SidePak™ AM510 Personal Aerosol Monitor manufactured by TSI Inc. was evaluated against a Tapered Element Oscillating Microbalance (TEOM) equipped with a Filter Dynamics Measurements System (FDMS). Conventionally, the SidePak is calibrated with respect to the Arizona Road Test Dust and then multiplied by an environmental calibration factor to yield mass concentration. To adapt this calibration to specific field conditions, we present an implementation of this calibration by introducing a growing factor correction which takes into account relative humidity and the dry and wet portions of the refractive index estimated from TEOM-FDMS measurements. PM2.5 sampling with several SidePaks AM510 was carried out in background and rural sites in the Po Valley (Italy). Modeled SidePak data were plotted vs. reference TEOM-FDMS data which show a good agreement.
Fetching data from CrossRef. This may take some time to load.
Journal of Environmental Monitoring
- Information Point